This work was part of a National Institute for Health Research participatory action research and practice development study, which focused on the use of a therapeutic, robotic baby seal (PARO, for personal assistive robot) in everyday practice in a single-site dementia unit in Sussex.
From the beginning of January 2017 until the end of September 2017, the cleaning and cleanliness of PARO was monitored through a service audit process that focused on the cleaning, amount of use and testing of contamination of PARO being used in everyday clinical practice with individuals and in group sessions. Its use and cleaning followed protocols developed by the study team, which incorporated hand hygiene and standard precaution policies. Its cleanliness was determined using an adenosine triphosphate (ATP) luminometer, with a benchmark of 50 relative light units (RLU). A reading of ATP below 50RLU is the level of cleanliness recommended for social areas in hospital settings. Throughout the study period, monitoring showed that all swab zones on PARO were within the benchmark of the 50RLU threshold for cleanliness.
PARO has an emerging evidence base as a useful therapeutic device. However, introducing such devices into clinical practice may encounter barriers or concerns from an infection prevention and control (IPC) perspective. This study of PARO in clinical practice aims to address the IPC concerns raised and offers cleaning and testing protocols and results.
Nursing Older People. 30, 3, 34-40. doi: 10.7748/nop.2018.e994Correspondence
This article has been subject to external double-blind review and has been checked for plagiarism using automated softwareConflict of interest
None declaredWrite for us
For information about writing for RCNi journals, contact email@example.com
For author guidelines, go to rcni.com/writeforus
Received: 07 September 2017
Accepted: 13 February 2018
Alternatively, you can purchase access to this article for the next seven days. Buy now